KD10 Series ODM Peristaltic Pump
Flow range: 0.0004~96mL/min
Max number of channels: 8
Send Email To Us
■ Lead Fluid KD10 series peristaltic pump flow range 0.0004~96ml/ min,multiple KD10 pump head can be up to 4-8 channels,flexible combination.
■ Parts adopt PVDF material ，with excellent chemical corrosion resistance.
■ 4 rollers and specially designed pressure tube structure make the liquid transfer more stable and lower pulsation.
■ Elastic pressure tube device provides accurate, smooth flow and excellent tube life.
■ Exquisite and beautiful appearance, compact structure, only small installation space required.
■ Matching high-quality steeper motor is an ideal choice precise dosing and discharging liquid to satisfied a wide range of instruments requirement.
Pump Head Flow Rate Parameters KD10\KD10-2
Tube material: Silicone, Pharmed,Viton,Tygon 3603 wall thickness 0.8~1mm
|Ml/min (Continuous working max100rpm)||0.4||1.8||5.8||16||38||48|
|Ml/min (Interval working max200rpm)||0.8||3.6||11.6||32||76||96|
★ Above parameters are obtained by using silicone tube to transfer pure water under normal temperature and pressure, in actually using it is effected by specific factors such as pressure ,medium. Above for reference only.
Pump Head Specifications
Channels: KD10-1 1 channel(maximum extensible to 4 channels) KD10-2 2 channels(maximum extensible to 2 channels)
Installation tube method: Whole(Tube Stopper Fixed)/ built-in connector(tube ID≥1.0mm)
Output pressure: ≤0.1Mpa
Net weight: 120g
Motor: stepper motor(42 type,57type)
42 type stepper motor
pump head: KD10-1, KD10-2
step angle1.8° \ phase2 \phase voltage6.44V \ phase current 0.7A \ phase resistance9.2Ω±10% \phase inductor16mH±20%
57 type stepper motor
pump head: KD10-1 (cascadable 1 to 4）, KD10-2(cascadable 1 to 4，speed＜200rpm）
step angle 1.8° \ phase 2 \phase voltage3.9V \ phase current 3A \ phase resistance1.3Ω±10% \ phase inductor 5mH±20%
Dimension (42 type stepper motor)
Dimension (57 type stepper motor)
Tube Lifetime Comparision Table
A.For longer tube life, choose thicker wall tube and low speed.
B.For larger flow,choose larger ID tube and in high speed.
C.For higher precision, choose smaller ID tube and medium to high speed
D.For higher suction and back pressure,choose thinner ID and thicker walltube and low speed.
Statement: The finalexplanation right of aboveinformation belongs to Lead Fluid.